What is it? Optical Tweezers use light to manipulate microscopic objects
as small as a single atom. The radiation pressure from a focused laser beam
is able to trap small particles. In the biological sciences, these instruments
have been used to apply forces in the pN-range and to measure displacements
in the nm range of objects ranging in size from 10 nm to over 100 mm.
Figure 1. Optical Tweezers principles.
How does it work? The most basic form of an optical trap is diagramed in Fig 1a. A laser beam is focused by a high-quality microscope objective to a spot in the specimen plane. This spot creates an "optical trap" which is able to hold a small particle at its center. The forces felt by this particle consist of the light scattering and gradient forces due to the interaction of the particle with the light (Fig 1b, see Details). Most frequently, optical tweezers are built by modifying a standard optical microscope. These instruments have evolved from simple tools to manipulate micron-sized objects to sophisticated devices under computer-control that can measure displacements and forces with high precision and accuracy.
Optical Tweezers have been used to trap dielectric spheres, viruses, bacteria, living cells, organelles, small metal particles, and even strands of DNA. Applications include confinement and organization (e.g. for cell sorting), tracking of movement (e.g. of bacteria), application and measurement of small forces, and altering of larger structures (such as cell membranes). Two of the main uses for optical traps have been the study of molecular motors and the physical properties of DNA. In both areas, a biological specimen is biochemically attached to a micron-sized glass or polystyrene bead that is then trapped. By attaching a single molecular motor (such as kinesin, myosin, RNA polymerase etc.) to such a bead, researchers have been able to probe motor properties such as: Does the motor take individual steps? What is the step size? How much force can the motor produce? Similarly, by attaching the beads to the ends of single pieces of DNA, experiments have measured the elasticity of the DNA, as well as the forces under which the DNA breaks or undergoes a phase transition.
Principle of Operation: Fig 1b shows a more detailed look at how an optical trap works. The basic principle behind optical tweezers is the momentum transfer associated with bending light. Light carries momentum that is proportional to its energy and in the direction of propagation. Any change in the direction of light, by reflection or refraction, will result in a change of the momentum of the light. If an object bends the light, changing its momentum, conservation of momentum requires that the object must undergo an equal and opposite momentum change. This gives rise to a force acting on the object.
In a typical optical tweezers setup the incoming light comes from a laser which has a "Gaussian intensity profile". Basically, the light at the center of the beam is brighter than the light at the edges. When this light interacts with a bead, the light rays are bent according the laws of reflection and refraction (two example rays are shown in Fig 1b). The sum of the forces from all such rays can be split into two components: Fscattering, the scattering force, pointing in the direction of the incident light (z, see axes in Fig 1b), and Fgradient, the gradient force, arising from the gradient of the Gaussian intensity profile and pointing in x-y plane towards the center of the beam (dotted line). The gradient force is a restoring force that pulls the bead into the center. If the contribution to Fscattering of the refracted rays is larger than that of the reflected rays then a restoring force is also created along the z-axis, and a stable trap will exist. Incidentally, the image of the bead can be projected onto a quadrant photodiode to measure nm-scale displacements (see Further Reading).
When the bead is displaced from the center of the trap, what force does it
feel? The restoring force of the optical trap works like an optical spring:
the force is proportional to the displacement out of the trap. In practice,
the bead is constantly moving with Brownian motion. But whenever it leaves the
center of the optical trap the restoring force pulls it back to the center.
If some external object, like a molecular motor, were to pull the bead away
from the center of the trap, a restoring force would be imparted to the bead
and thus to the motor. An example trace of a single kinesin motor taking 8 nm
steps against a 5-pN force is shown in Fig 2.
Figure 2. Measurement of the 8-nm steps of kinesin.
Modern Optical Tweezers: In practice, optical tweezers are very expensive, custom-built instruments. These instruments usually start with a commercial optical microscope but add extensive modifications. In addition, the capability to couple multiple lasers into the microscope poses another challenge. High power infrared laser beams are often used to achieve high trapping stiffness with minimal photo-damage to biological samples. Precise steering of the optical trap is accomplished with lenses, mirrors, and acousto/electro-optical devices that can be controlled via computer. Figure 3 is meant to give an idea of the number of elements in such a system. In short, these are very complicated instruments that require a working knowledge of microscopy, optics, and laser techniques.
Figure 3. Typical modern optical tweezers setup.
A. Ashkin, J.M. Dziedzic, J.E. Bjorkholm and S. Chu. 1986. "Observation of a Single-Beam Gradient Force Optical Trap for Dielectric Particles." Opt. Lett. 11 (5) 288-290.
1. Block S. M. 1992. "Making light work with optical tweezers." Nature
2. Svoboda K., Block S. M. 1994. "Biological applications of optical forces." Annu Rev Biophys Biomol Struct 23:247-85.
3. Simmons R. M., Finer J. T., Chu S., Spudich J. A. 1996. "Quantitative measurements of force and displacement using an optical trap." Biophys J 70(4):1813-22.
4. Visscher K., Gross S. P., Block S. M. 1996. "Construction of multiple-beam optical traps with nanometer-resolution position sensing." IEEE Journal of Selected Topics in Quantum Electronics 2(4):1066-1076.
5. Gittes F., Schmidt C. F. 1998. "Interference model for back-focal-plane displacement detection in optical tweezers." Optics Letters 23(1):7-9.
6. Peters I. M., de Grooth B. G., Schins J. M., Figdor C. G., Greve J. 1998. "Three dimensional single-particle tracking with nanometer resolution." Review of Scientific Instruments 69(7):2762-6.
7. Smith S. P., Bhalotra S. R., Brody A. L., Brown B. L., Boyda E. K., Prentiss M. 1998. "Inexpensive optical tweezers for undergraduate laboratories." Am. J. Phys. 67(1):26-35.
8. Visscher K., Block S. M. 1998. "Versatile optical traps with feedback control." Methods Enzymol 298:460-89.
9. Neuman K. C., Chadd E. H., Liou G. F., Bergman K., Block S. M. 1999. "Characterization of photodamage to escherichia coli in optical traps." Biophys J 77(5):2856-63.
10. Pralle A., Prummer M., Florin E. L., Stelzer E. H., Horber J. K. 1999. "Three-dimensional high-resolution particle tracking for optical tweezers by forward scattered light." Microsc Res Tech 44(5):378-86.
11. Ashkin, A. 2000. "History of Optical Trapping and Manipulation of Small-Neutral Particle, Atoms, and Molecules." IEEE Journal of Selected Topics in Quantum Electronics 6(6):841-856.
12. Ishii Y., Ishijima A., Yanagida T. 2001." Single molecule nanomanipulation of biomolecules." Trends Biotechnol 19(6):211-6.
13. Kuo S. C. 2001. "Using optics to measure biological forces and mechanics." Traffic 2(11):757-63.
14. Lang M. J., Asbury C. L., Shaevitz J. W., Block S. M. 2002. "An automated two-dimensional optical force clamp for single molecule studies." Biophys J 83(1):491-501.
Atomic Force Microscopy, Magnetic Tweezers